|(800) 853-6785 · (919) 732-5400 · FAX (919) 732-6153 · firstname.lastname@example.org|
The MI-5000B Series II Microprocessor Audiometer is one of the lowest-priced field serviceable microprocessor audiometers available today. It administers computerized tests securing valid, reliable data quickly and cost effectively. It is very easy to understand and use, and can be used either alone, or interfaced with an external printer or computer. Interface settings can be changed by the operator. Tones are generated and controlled entirely by digital circuits, allowing output levels of the MI-5000B to be adjusted by through secured keyboard entry. The LCD display simultaneously shows all thresholds established for both ears, as well as on-going test conditions.
Test Frequencies: 500, 1000, 2000, 3000, 4000, and 6000 Hz. 8000 Hz is optional.
Frequency Accuracy: Crystal controlled, less than 1% error at all frequencies.
Frequency Sequence: Start either ear, 1000, 500, 1000 retest, 2000, 3000, 4000, 6000, (optional 8000). Second ear, 500, 1000, 2000, 3000, 4000, 6000 (and 8000 optional).
Intensity Range: From 0 dB to 90 dB Hearing Level (HL) in 5 dB steps.
Attenuator Linearity: Less than 0.75 dB error for any 5 dB step, less than 1 dB error for any 10 dB step, less than 2 dB accumulated error relative to the calibration level.
Tone Rise/Fall Times: 44/32 ms typical.
Test Paradigm: Modified Hugheson Westlake in the automatic mode with fully manual override capability.
Testing Time: On a cooperative individual or a biological simulator, approximately 5 ½ minutes (testing 8 kHz.).
Error Warnings: Audible alert (when active) with visual display of specific
error encountered. The following error conditions are detected and signaled.
Audiometer Calibration: All audiometer calibration parameters will meet ANSI S3.6 1996 Standard for Audiometers and OSHA 29 CFR 1910.95. Output levels are calibrated through secured keyboard entry.
Safety: Listed to UL544 Standard for Medical and Dental Equipment and CSA C22.2 No. 125-M1984 Environment Products/Health Care Technology.
Earphone: Telephonics Corporation TDH -49 earphones with MX-41/AR cushions.
Power Requirements: 120 VAC, 60 Hz, 32 VA
Physical Dimensions: 2.75" high, 7.5" wide, 9" deep.
Net Weight: Unit, 2.6 lbs; Earphone, 1 lb; Transformer, 1.2 lbs.
Monitor Instruments, Inc. 437 Dimmocks Mill Road, Suite 50, Hillsborough, NC 27278
(800) 853-6785 (919) 732-5400 FAX (919) 732-6153 info@MonitorInstrumentsInc.com
Copyright © 2001, Monitor Instruments, Inc. All Rights Reserved